University of Waterloo >
Electronic Theses and Dissertations (UW) >
Please use this identifier to cite or link to this item:
|Title: ||Protein sorption to contact lenses and intraocular lenses|
|Authors: ||Luensmann, Doerte|
|Keywords: ||protein sorption|
confocal laser scanning microscopy
|Approved Date: ||22-Dec-2009 |
|Date Submitted: ||2009 |
To locate protein sorption on the surface and inside the matrix of soft contact lens materials and intraocular lenses (IOL).
The proteins albumin and lysozyme were investigated as they are highly abundant in blood serum and tears, respectively. Proteins were conjugated with organic fluorescent probes and using confocal laser scanning microscopy (CLSM) the sorption profile to contact lenses and IOL could be determined. Radiolabeled protein was used for quantification purposes.
• Albumin sorption to etafilcon A and lotrafilcon B was determined (Chapter 3)
• Different fluorescent probes were used for conjugation and the impact on albumin sorption behaviour was investigated (Chapter 4)
• Lysozyme sorption to nine different pHEMA-based and silicone hydrogel contact lenses was determined using two fluorescent probes (Chapter 5)
• The efficiency of protein removal from contact lenses using contact lens care regimens was investigated (Chapter 6)
• Albumin sorption to IOL materials was quantified and imaged using a modified CLSM technique (Chapter 7)
Albumin and lysozyme sorption profiles differed between materials, and were influenced by the fluorescent probes used for conjugation. After one day of incubation, both proteins could be located within all contact lens materials, except for lotrafilcon A and lotrafilcon B, which primarily allowed deposition on the lens surface. An increase in protein accumulation was found for most materials over the maximum investigated period of 14 days, using CLSM and radiolabel techniques. The efficiency of contact lens care regimens to remove lysozyme and albumin depended on the lens material, care regimen and protein type investigated.
PMMA and silicone IOLs showed protein exclusively on the surface, while a hydrophilic acrylic IOL allowed penetration into the lens matrix over time. Despite the albumin penetration depth into hydrophilic acrylic, the highest albumin levels were determined for the silicone IOL.
CLSM provides detailed information that can describe the protein distribution in transparent biomaterials, with scanning depths up to a few hundred microns. However, the CLSM data are primarily of qualitative value, which necessitates a quantitative technique (e.g. radiolabeling) to determine the total protein content.|
|Program: ||Vision Science|
|Department: ||School of Optometry|
|Degree: ||Doctor of Philosophy|
|Appears in Collections:||Electronic Theses and Dissertations (UW)|
Faculty of Science Theses and Dissertations
All items in UWSpace are protected by copyright, with all rights reserved.